Published on December 28, 2008
Notes to a Low Temperature Physicist or, what I wished I'd have known when I started LT Techniques course, 26th Nov 2008 Dr Kurt Haselwimmer Cambridge Magnetic Refrigeration
PhD = 1000 days
LT is difficult
A Cabinet of Curiosities ? Ole Worm (1588-1654)
Another Curiosity ? http://www.flickr.com/photos/noyespictures/228951299
Pulse tubes now to 2.7K Dryogenic – DMS-1000 system
http://www.flickr.com/photos/andrewmckaskill/16531297 “God hangs the greatest weights upon the smallest wires” Francis Bacon
Don’t build if you can buy
http://www.cmr-direct.com Rewiring a cryostat properly costs less than getting a power socket installed. Do it right and do it only once.
Pumps and plumbing
http://www.flickr.com/photos/arcanegazebo/445426559 / “A vacuum is a hell of a lot better than some of the stuff that nature replaces it with” Tennessee Williams
• Always clean and grease o-rings • Only Use Viton O-rings – buna-N cracks • Paint your KF-fittings • Don't use poppet relief valves on vacuum cans • Always boil-off flux on HS joints • Buy SS flexibles from Kurt J Lesker • Use plastic KF clamps from BOC edwards • Use only bellows sealed valves on He-3
http://www.flickr.com/photos/arcanegazebo/3299089 The dilution fridge – the mainframe of mK machines
Keep it simple, stupid !
Simplicity means achieving the maximum effect with the minimum means. Koichi Kawana http://www.flickr.com/photos/hynkle/2587828754
Entropy, S Temperature
Entropy, S B min = 0.1 T B max = 6T Temperature T MIN <50mK T BATH = 4.2K, 1.5K
Entropy, S B min = 0.1 T B max = 6T Temperature
Superconducting Magnet Refrigerant
Entropy, S B min = 0.1 T 1 B max = 6T Temperature T BATH = 4.2K, 1.5K
Bath Thermal Contact Superconducting Magnet Refrigerant Pill
Entropy, S B min = 0.1 T 1 B max = 6T 2 Temperature T MIN <50mK T BATH = 4.2K, 1.5K
Bath Heat Switch Superconducting Magnet Refrigerant Pill
Connection to Bath Heatswitch Sample Region (Double axis rotator) Refrigerant Practical Refrigerators
a LT is difficult
Cambridge Magnetic Refrigeration www.cmr.uk.com Cryoconsumables - wiring, connectors etc www.cmr-direct.com Completely Dry systems -Dryogenic Ltd www.dryogenic.com Dr Kurt Haselwimmer Kurt@cmr.uk.com
Presentación que realice en el Evento Nacional de Gobierno Abierto, realizado los ...
In this presentation we will describe our experience developing with a highly dyna...
Presentation to the LITA Forum 7th November 2014 Albuquerque, NM
Un recorrido por los cambios que nos generará el wearabletech en el futuro
Um paralelo entre as novidades & mercado em Wearable Computing e Tecnologias Assis...
Low Temperature Techniques Course. ... students whose research requires the ability to perform experiments at low temperatures. ... firstname.lastname@example.org.
Low Temperature Techniques Course. ... Organised by the IOP Low Temperature Group. ... There is also an introduction to commercially available low ...
Published 19 February 2008 Online at stacks.iop.org/JPhysCM/20/104246 ... high-pressure and ultra-low-temperature techniques must be ... In the course of
... Matter Physics At Low Temperatures ... 2008 Mercury Mariner Owners ... techniques in condensed matter physics at low temperatures advanced ...
... We use holographic techniques to study meson quasiparticles moving ... low temperature phase is that moving through the thermal ... Of course, this ...
Clinical cardiac electrophysiology ... Established technique for ~ 30 ... temperature (only in solid tip catheters)
March 2008 . iii ... waste heat. However, during the course of this study, ... Lowtemperature heat recovery faces at least three challenges:
PUBLISHED BYIOP P UBLISHING TOP PAPERS 2008 SHOWCASE ... Of course, these papers appear here, and are just two of the papers which deserve the heightened
Forthcoming volumes. Some of our forthcoming volumes include: XIV International Conference on Topics in Astroparticle and Underground Physics (TAUP2015), Italy